Why would you want to measure mass and thickness of thin films? One reason to monitor these two parameters is to characterize the build-up and degradation of molecular layers. Also, numerous materials and thin coatings are dynamic in nature and undergo physical changes responding to various stimuli, such as light, temperature, salt concentration or pH. These changes influence the mass, thickness and the structural properties of the material. The mass and thickness are therefore two parameters that are very much involved both the creation, behavior, and degradation of thin films. This makes them relevant to monitor both in the design, characterization, evaluation and optimization of thin films and coatings.
During surface interaction processes such as for example binding of molecules, adsorption, desorption, aggregation, and build-up of multilayer films, the mass and thickness of the molecular layers change. By measuring these changes in real time, we can follow the molecular binding processes and rearrangements. To measure these changes on a molecular scale, we need real-time nanoscale techniques. Once such technique is Quartz Crystal Microbalance with Dissipation monitoring technology (QCM-D). The QCM-D measures the changes of two parameters, the resonance frequency (f) and the energy dissipation (D). From these two parameters, mass and thickness changes at the surface can be extracted. In general, as the mass increases at the surface, the f will decrease. The D-parameter will indicate how soft the layer is. The softer the layer the higher the D. In the case of mass loss, the frequency will instead increase. And if the layer goes from soft to stiff, then the D parameter will decrease.
Molecules that are typically studied with this surface sensitive technology include lipids, proteins, DNA, polymers, surfactants, nanoparticles.
Let us take an example of a mass measurement. Here we are interested in detecting when an anti-biotin antibody first binds to a biotinylated lipid bilayer. We would also like to detect the cleavage of the antibody by an enzyme.
This measurement shows how monitoring of the mass changes allows us to detect both the binding of the antibody and the cleavage by the enzyme. Overall, the measurement offers insight in to both the molecular interaction processes and verifies the function of the enzyme as well as the direction of the antibody at the surface.
Figure 1. Mass changes as an antibody binds to the biotinylated bilayer on the surface (I-II) followed by enzyme cleavage (III) which removes 1/3 of the mass at the surface.
Download our overview to read more about what information you can extract from QCM-D analysis.
Recent research results could help designing formulations better stabilized against protein aggregation, one of the main challenges in pharmaceutical development. QSense QCM-D analysis helped piecing the puzzle together
Read about how and why the QCM fundamental frequency matters in measurements
Read about how QSense technology is used to analyze contaminant adsorption and desorption, and to compare the efficiency of post CMP-cleaners.
Read about how polyelectrolyte multilayer build-up, and the resulting multilayer, can be characterized with QSense QCM-D technology.
The fundamental resonance frequency of QCM-sensors is often 5MHz. Learn about the theory behind the number.
Learn about how QSense® technology can be used to analyze CMP-related processes, such as additive - surface interaction, residue removal, and etching.
To prepare for the next pandemic, we need to think outside the box. One approach could be broad-spectrum antiviral targeting.
Read about why it is possible to gain valuable information from a viscoelastic sample by monitoring multiple overtones in QCM measurements.
The D-factor provides information that is complementary to the frequency response. Read about how it can be understood and what information it reveals.